# Modulus of Elasticity

## What is Modulus of Elasticity?

Modulus of Elasticity, also known as Elastic Modulus or simply Modulus, is the measurement of a material's elasticity. Elastic modulus quantifies a material's resistance to non-permanent, or elastic, deformation. When under stress, materials will first exhibit elastic properties: the stress causes them to deform, but the material will return to its previous state after the stress is removed. After passing through the elastic region and through their yield point, materials enter a plastic region, where they exhibit permanent deformation even after the tensile stress is removed. In the graph below, the elastic region is highlighted in yellow and the plastic region is highlighted in green.

Modulus is the slope of the straight-line portion of a stress (σ) strain (ε) curve. Focusing on the elastic region, if the slope is taken between two stress-strain points, the modulus is the change in stress divided by the change in strain. Modulus =(σ2 - σ1) / (ε2 - ε1) where stress (σ) is force divided by the specimen's cross-sectional area and strain (ε) is the change in length of the material divided by the material's original gauge length.

Since both stress and strain are normalized measurements, modulus is a consistent material property and can be compared regardless of material size. That is, a large steel specimen will have the same modulus as a tiny steel specimen, although the large specimen will require a higher maximum force to deform the material. Since modulus is simply the slope of the stress/strain curve, materials with a steeper slope are stiffer, with a higher elastic modulus than a material with a shallow slope. Although modulus is most commonly reported during a tensile test, modulus of elasticity can also be reported as compressive modulus of elasticity for compression tests,  flexural modulus of elasticity for flexural tests, shear modulus of elasticity for shear tests, or or torsional modulus of elasticity for torsion tests. Modulus of elasticity may be determined by dynamic testing, where it can be derived from complex modulus.

## Calculating Different Types of Modulus

Users recording modulus should be aware that there are many different ways to calculate the slope of the initial linear portion of a stress/strain curve. For example, Instron's Bluehill Universal software offers more than eight different ways to calculate modulus of elasticity. When comparing modulus results for a given material between different laboratories, it is critical to know which type of modulus calculation was chosen. Below is a sample of three different modulus calculations.

### Chord Modulus

A user selects a start strain point and an end strain point. A line is drawn between the two points and the slope of that line is recorded as the modulus.

### Secant Modulus

Uses the zero stress/strain point as the start value and a user-selected strain point as the end value. A line is drawn between the two points, and the slope of that line is recorded as the modulus.

### Segment Modulus

A user selects a start strain point and an end strain point. Using the least-squares fit on all points between the start and end points, a line is constructed. The slope of the best fit line is reported as the modulus.

### Tangent Modulus

A user selects a tangent point on the stress/strain curve. The slope of the tangent line is reported as the modulus.